The SAVer cables are designed to ‘save’ the instrument front panel from dirt, scratches and potentially expensive repairs and downtime. When the SAVer cable is used with the high quality mating adapter, MAAD-FC, your front panels will stay clean without adding loss. The SAVer cable can be repolished or replaced quickly and easily.
Instrument End: Connect SAVer cable to instrument output. DO NOT DISCONNECT THIS END AFTER THIS STEP.
Reference Cable End: Connect Reference Cable to SAVer cable using only high quality bulkheads like the MAAD-FC mating adapter.
Note: Be sure all optical interfaces are clean and inspected (400x microscope) before mating.
Once connected to the instrument, this SAVer cable would not be disconnected for normal referencing and testing.
Note: Clean and inspect SAVer cable with every mating.
Next, connect three meter HPR reference cable to the SAVer cable using MAAD-FC mating adapter.
Now you can begin your referencing process.